Press Autoscale to activate its menu. Portable mixed signal digital storage oscilloscope pages. The reading shows the trigger source. The display in the screen is shown as Fig. Press F3 next to Do Self-Cal. It has European Conformity CE marking.
You must be logged in to post a review. Channel 2 as the trigger source. You can match these frequencies with known system frequencies, such as system clocks, oscillators, or power supplies. They may provide real-time intensity grading, and may have the ability to adjust frequency of display. Oscilloscopes can be powered with batteries or electricity, or may have a built-in generator. There are three kinds of trigger modes available for this oscilloscope, which is, auto, normal and single.
They are commonly used for testing, measuring, and inspecting in science, engineering, telecommunications, automotive, and electronics applications.
Some oscilloscopes can capture waveform signals to memory, and recall them. The purple pointer shows the trigger level position.
Owon PDST – Specs, Manuals & Buy
Warning indicates the conditions or practices that could result in injury or loss of life. HF Rjc Block the high-frequency signal and only unblock Coupling the low-frequency component. Don’t show me this message again. In order to avoid any contingent danger, this product is only used within the range specified. The following functions would be disabled when XY Format is set.
The screen display is shown as Fig. Page 3 User Manual of PDS Series Oscilloscope General Warranty The Lilliput warrants that the product will be free from defects in materials and workmanship for a period of three years from the date of purchase of the product by the original purchaser from the Lilliput company. Page 8 User Manual of PDS Series Oscilloscope To avoid body damage and prevent product and connected equipment damage, carefully read the following safety information before using the test tool.
Check whether the power switch is pushed down to the designated position.
Page 62 – Example 6: Please charge the battery for 12 hours to make sure enough power to supply the oscilloscope should be turned on during charging to oscilloscope. Press F3 next to Do Self-Cal. The company, founded in in China, has a U.
Press the F1 menu selection button and choose Edge as the trigger type. The following terms may appear in this manual: Press the F4 menu selection button and choose Single as the trigger mode. Each trigger mode has its function menu for users to set them up. If it is required to change check the set value of the probe attenuation coefficient, press the function menu button of the channels used, then push down the selection button corresponding to the probe till the correct digitap value is shown.
Measuring a Simple Signal The purpose of this example is to measure the frequency and peak-to-peak value of a given signal.
Bandwidth is the maximum frequency of a signal that an oscilloscope can capture. The waveform shows the sweeps or refresh of voltage on a vertical Y axis, and the time on a horizontal X axis.
Portab,e routine uses internally generated signals to optimize circuits that affect channel sensitivity, offset and trigger parameters.
Taking the additive operation between Channel 1 and Channels 2 for example, the operation steps are as follows: Details of a waveform demonstrate the signal over time.
Reviews There are no reviews yet. In order to avoid suffering from the electric shock, please keep your finger behind the safety guard ring of the probe body sefies the operation. Add and subtract math functions plus FFT fast Fourier transform along with automatic measurements such as period, frequency, and peak-to-peak enable an in-depth analysis of waveforms.
It shows the selected trigger type: A constant image will display if a signal is shown at a high frequency.
Check whether there is any damage caused by transportation. Video Pds50022t Trigger Observe the video circuit of a television, apply the video trigger and obtain the stable video output signal display. The square wave of 1 KHz frequency and 5V peak-peak value will be displayed in several seconds see Fig.
Function Setting Description Menu Channel 1 as the trigger source. Only if the oscilloscope fulfill the following two conditions at first, can these specification standards be reached. However, OWON will continue to improve products and reserves the rights to changes specification at any time without notice.
RS data line or USB data line: Oscilloscopes can be powered with batteries or electricity, or may have a built-in generator. The reading shows the trigger source. Press F3 next to Sync and select Field. It has European Conformity CE marking. Press the F2 to choose CH1 for trigger source.
The probe which is not compensated or presents a compensation deviation will result in the measuring error or mistake. The most negative peak voltage measured over Vmin the entire waveform. Recall Factory Reset system settings to factory default.
Source Use external source.